[SEN-2514] During a 5G OTA measurement, if the signal loses lock and then regains lock the graph doesn’t update
[SEN-2476] During an OTA measurement, the widget was indicating the opposite sign to the graph
Note: This version of firmware can only be installed on Sentinel instrument with a serial number greater than 00300000.
HTTP(S) Interface An HTTP(S) interface has now been added to Sentinel allowing a basic remote control functionality. For full details please refer to the Sentinel User Manual.
Continuous Measurements An option has now been added to the measurement settings page to allow the user to select continuous measurements. This means that Sentinel will continue to make a measurement on a rolling 24 hr basis.
IPv6 on the Management Port There is now the ability to use IPv6 addresses on the management port for Sentinel.
Disable SyncE SyncE can now be disabled independently of PTP/NTP measurements. This feature is available via the mode page.
Disable Metrics If you don’t want to run any metrics or masks you can disable this on the measurement settings page.
New options and hardware are required to enable 10GbE functionality. Existing instruments with serial numbers greater than 300000 may be hardware-upgraded for 10GbE functionality; this upgrade is return-to-factory.
CAT v24.00 or later is required for data analysis
Non-10GbE instruments may also be upgraded to R11
[SEN-1085] Ethernet Gateway Ping button is enabled when DHCP is enabled
[SEN-1151] Sentinel doesn't reach 128 packets per second for Delay Request rate
[SEN-1726] Configuring Sentinel from Electrical to optical ethernet results in no RevPDV being recorded
[SEN-1781] Apply button will only change IP address / Mask / Gateway setting on the first press
[SEN-1796] Protocol check does not detect Delay Response if multiple PTP multicast flows are present
[SEN-1803] Moving from GNSS settings page with invalid value on antenna delay causes crash